Manual was delivered in a timely manner and was all in English as advertised. The manuals I received when we moved into our flat were in German, Italian, and French. Having never used a steamer before, and not speaking/reading German very well, I needed an English Manual. this was a huge help.
This GRUNDIG UV5A Owner Manual is not only an instruction manual but a total functional circuit description of the Multimeter and includes circuit diagram at the end pages. It is very helpful for repairing and calibrating the instrument. It is written in two languages English and German for international support. It was very easy to repair my unit with this document. Regards, Regis Pauly, Electrical Engineer.
The owner's manual was evetything I thought it would be. I was able to print it out using both sides of the paper in quick order, saving me a search online for a used one.
Text excerpt from page 51 (click to view)
The unit can be mounted into a 19-inch standard rack if the optional rack-mounting adaptors (AJ-MA75P) are used. For the installation rails, it is recommended that the rail and bracket for 18" length (model number CC3001-99-0400) of CHASSIS TRAK be used. (The complete slide rail and bracket unit is not available from Panasonic) For further details, consult with your dealer.
Remove the screws on the left and right sides of the unit. Use the removed screw to attach the inner members of the slide rails.
The length of the screws used is subject to restriction. If some of the mounting screws have been lost or misplaced, use screws which are less than 4" long in their place. Use four screws to secure each inner member.
Attach the outer member brackets to the rack. Check that the height is the same for the left and right brackets. Attach the AJ-MA75P rack-mounting adaptors with included 4 screws.
Remove the 4 rubber legs from the bottom of the unit, and install the unit in the rack. After the unit has been installed, check that it moves smoothly along the rails.
EIA standard rack Fasten it to the rack with set screws.
<Notes> â¢ Keep the temperature inside the rack to between +41Â°F (5Â°C) and +104Â°F (40Â°C). â¢ Bolt the rack securely to the floor so that it will not topple over when the VTR is drawn out.