There are currently no product reviews.
Great to have extra resources for Service Manuals, Now days you can really not trouble shoot efficiently without one , Wayne at IRIONS TV & ELECTRONICS REPAIR Clearwater , Fl. 33755 727-446-7955
For five bucks you can barely buy a hamburger. Or for the same five bucks you can buy a service manual. Much more useful. (and better for your health, depending on where you buy your hamburgers).
Yes, there are free manual sites out there, but if they don't have what you want, you have to pay.
And five bucks isn't much. Not for full specs, schematics and adjustment and parts replacement procedures.
My only criticism is that grayscale illustrations aren't well rendered, but I've seen worse.
Schematics and text are clear.
I'll be happy to purchase from here again.
Impressively thorough. Even the simple operators manual helped me "fix" one of the 2 CD players in the unit. This unit reads CD's from the top so they should be installed in the magazines "upside down" from typical CD players. The CD player service manual helped me unjam a stuck carriage because somebody transported the box laying down loaded with CD's. A little lens cleaning & the player now works well! Thanks for you help at a great price! Joe
I was skeptical at first but later found the manual to be good quality for the price. It took a couple hours to receive the email with the download link, well worth the wait. Thanks.
very helpful, I could not have cleaned motherboard and replaced the main fan without it
NN-T793SF / NN-T783SF / NN-T763SF / NN-S763BF / NN-S763WF / NN-S753BF / NN-S753WF / NN-S743BF / NN-S743WF
4 PROCEDURE FOR MEASURING MICROWAVE ENERGY LEAKAGE
4.2.1. Measurement with the outer panel removed.
Whenever you replace the magnetron, measure for radiation leakage before the outer panel is installed and after all necessary components are replaced or adjusted. Special care should be taken in measuring around the magnetron.
NOTE: The U.S. Government standard is 5 mW/cm 2 while in the customerâs home. 2mW/cm 2 stated here is our own voluntary standard. (1mW/cm2 for Canada)
Measurements with a fully assembled oven.
After all components, including outer panel are fully assembled, measure for radiation leakage around the door periphery, the door viewing window, the exhaust opening and air inlet openings.
â¢ Electromagnatic radiation monitor â¢ 600cc glass beaker
â¢ Glass thermometer 212Â°F or 100Â°C
Record keeping and notification after measurement
Procedure for measuring radiation leakage
â¢ After any adjustment or repair to a microwave oven, a leakage reading must be taken. Record this leakage reading on the repair ticket even if it is zero. A copy of this repair ticket and the microwave leakage reading should be kept by repair facility. â¢ Should the radiation leakage be more than 2 mW/cm2 (1mW/cm for Canada) after determining that all parts are in good condition, functioning properly, and genuine replacement parts as listed in this manual have been used, immediately notify PASC, PSC or PCI.
Note before measuring. â¢ Do not exceed meter full scale deflection. Leakage monitor â¢ should initially be set to the highest scale. â¢ To prevent false readings the test probe should be held by the grip portion of the handle only and moved along the shaded area in Figure no faster than 1 inch/sec (2.5cm/sec). â¢ Leakage with the outer panel removed ...... less than 5mW/cm2. â¢ Leakage for a fully assembled oven with door normally closed ...... less than 2mW/cm 2 (1mW/cm 2 for Canada). â¢ Leakage for a fully assembled oven [Before the latch switch (primary) is interrupted] while pulling the door ...... less than 2mW/cm 2. 1. Pour 275 Â± 15cc (9ozssÂ± 1/2oz) of 20Â°C Â± 5Â°C (68Â° Â± 9Â°F) water in a beaker which is graduated to 600cc, and place in the center of the oven. 2. Set the radiation monitor to 2450MHz and use it following the manufacturerÂ´s recommended test procedure to assure correct results. 3. When measuring the leakage, always use the 2 inch (5cm) spacer supplied with the probe. 4. Tap the start pad or set the timer and with the magnetron oscillating, measure the leakage by holding the probe perpendicular to the surface being measured.
At least once a year, have the radiation monitor checked for calibration by its manufacturer.